Cell Disruptor Transducer Ultrasonic Cell Disruptor For Smash Plant Cells
The ultrasonic cell disruptor is made up of two parts: ultrasonic generator and transducer,
the former transiting 50KHz 220V into 20KHz as power supply to the latter.
Zirconiumtitanium acid barium piezoelectric vibrator is the heart of transducer,
which works telescopic elastic deformation at a frequency of 20KHz according to
alternating voltage andaccordingly,the transducer responds by portrait mechanical vibration.
The vibration wave via a Ti alloy amplitude bar,
which is soaked in the biological solution, produces cavitation effect and
dries biological particles in the medium librating hard.
The ultrasonic cell disruptor can prepare emulsoid of 1/100 micron,
homogenize mixture hard to be mixed polymerize some matters and separate out some other.
Generally speaking, it can process various functions such as distillation, smash, emulsification,
homogenization, soliquoid, aberrance, aerosol,quicken up exsolution and crystallization
and preparing different kinds of biological samples under an electron microscope.
Disposing of organic and inorgani compound safely and efficiently Mainly used for crush or
virus and bacterium .
Mainly used for crush of virus and bacterium of the cell and tissue of the various of animal or plant,
crush and combination again of the various of inorganic compound.
Used in emulsification,separation, even,recovering and chemical reaction acceleration etc.
Widely used in the field of biochemistry,microbiology, pharmacology,medicine and etc.
1. The digital display and control, convenient and accurate.
2. Regular time setting button (4), the left two keys to set minutes of increase and decrease,
the right of the two key set seconds of increase and decrease
3.The "Pulse" : works display window. In the intermittent work way is divided into two parts:
the left two shows that intermittent way. Pulse width, the range of 0.1 ~ 9.9 seconds,
by ", "said. In "continuous" works shows the "Cont."
4. The intermittent work way of "duty cycle" set button.
Two key set for intermittent mode pulse width on the left.
Two key way of intermittent intermittent time setting key on the right.
5.Mode ": continuous output work Mode or" pulse output "work Mode selection button.
Remark: use the intermittent work way can prevent the accumulation of excessive heat in the liquid,
and because the material to be able to accumulate after each pulse cluster probe oscillator,
so can provide a more effective treatment results.
6. “Start/Stop” Work start or stop button.
7. “Intensity display”:The intensity of ultrasonic output display window, display range of 0 ~ 100%.
8. “Max.intensity”:Ultrasonic output intensity limit knob, a total of 4, 20%, 35%, 60% and 100% respectively.
Note: as a strong power output will damage the amplitude of small diameter,
so please select a location according to the following requirements:
9. “Intensity” :Ultrasonic output intensity, can be in "the intensity of ultrasonic output restrictions" knob
selected range from 0 ~ 100% range adjustment.
10. “Tune” :Ultrasonic output frequency tracking fine tuning hole,
generally do not need to adjust, but after the user to replace the amplitude may need to be adjusted a little.
11“Stand by”: Overheating protection lamp, when the temperature sensor is detected
by the temperature too high, the indicator light will flash, the host will cease to work,
the normal work of the host, the indicator light is not bright.
12: "POWER" : the POWER switch, from one side of the "ON" to connect the POWER,
ON the other side to shut off the POWER.
Type Power(w ) Optional luffing rod Handle Sample Weight Plus duty cycle work
UH-100A 100 1/8″3/16″1/4″ 100ul~ 80ml 1%~99%
UH-250A 250 1/8″3/16″1/4″ 100ul~ 200ml 1%~99%
UH-500A 500 1/8″3/16″1/4″1/2″ 250ul~ 500ml 1%~99%
UH-800A 800 3/16″1/4″1/2″3/4″ 250ul~ 800ml 1%~99%
UH-950A 950 3/16″1/4″1/2″3/4″ 250ul~ 1000ml 1%~99%
UH-1200A 1200 3/16″1/4″1/2″3/4″ 50ml~ 1200ml 1%~99%